Volume-weight is one of the required inspection items in food acquisition. Volume-weight tester is developed according to the standard of LS/T3701-1993 "Of HGT-1000 grain bulk density", and the measurement of the grain volume-weight in GB1353-2009 "corn" new GB.
Electronic Volume-weight Apparatus is widely used in volume-weight testing of grain such as wheat, rice, sorghum, corn and so on . It can work out the levels according to standard requirements at the same time. Therefore, it plays a key role in grain purchasing, storage, processing
2. Adopts imported patch components with high density and high reliability instead of traditional electronic components, which saves the components installing space of inner structure.
3. Adopts advanced electronic weighing technology and computer technology, and has realized displaying weighing results meanwhile automatically printing rest results.
4. The machine parts has been selected and processed delicately, with unique color and beautiful outlook.
5. Easy to operate, with high accuracy and fast measurement.
Max weighting:1000±2g Min weighting:100g Resolution:1g Volume-weight cylinder volume:1000±1.5ml Working voltage:220V Power:10W Working environment requirements: Environment temperature:5-40°C Relative humidity:<90%RH
Small grain electronic volume-weight apparatus
Small grain electronic volume-weight apparatus(with printing function)
Large grain electronic volume-weight apparatus
Large grain electronic volume-weight apparatus(with printing function)